Bamboo Settings for Wagner Meters Handheld Moisture Meters
Settings* for Johnson Premium Hardwood Flooring Bamboo Flooring Products
|MATERIAL DESCRIPTION||JOHNSON PART NO.||SETTING|
|Horizontal Bamboo (natural)||JVC-9601||0.59|
|Vertical Bamboo (natural)||JVC-9603||0.58|
|Vertical Bamboo (carbonized)||JVC-9604||0.53|
|Strand Bamboo (natural)||JVC-9605||0.76|
|Strand Bamboo (carbonized)||JVC-9606||0.81|
Settings* for Teragren® Bamboo Flooring Products
|MATERIAL DESCRIPTION||TERAGREN® PART NO.||SETTING|
|Stranded Wheat-Color, Not Carbonized||TPF-SYN-WHT-MPL||0.97|
|Traditional (not-stranded) Flat-grain, carbonized||PF-FGC-MPL||0.67|
|Stranded, Chestnut, carbonized||TPF-SYN-CHSNT-MPL||1|
|Traditional (not-stranded) Vertical-grain, carbonized||PF-VGC-MPL||0.66|
|Traditional (not-stranded), Flat-grain, not-carbonized||PF-FGN-MPL||0.69|
*These settings were derived from sample populations of the different flooring products provided by the above manufacturers. These samples were lab-tested for actual moisture content at Wagner Meters using the ASTM D4442-92 oven-dry method. The oven-dry results were then used to establish the optimum settings for Wagner Meters digital handheld moisture meters.
ASTM D4442-92 is the laboratory standard method used to verify the level of accuracy of handheld moisture meters for wood.
Decimal point is not shown on the meter display. Example: when setting your meter for .59, the actual value on the meter display will be 59.