When I use my Rapid RH® Easy Reader, all I see are 2 lines. What should I do?
If the reader is displaying two lines (–) then immediately back it out of a Smart Sensor. Check to see if this problem occurs with multiple sensors (not just related to an individual sensor). If the problem occurs with another Smart Sensor, then most likely the problem is that the connector at the bottom of the Easy Reader is not making proper contact with the sensor(s). Inspect the connector to see if any of the leaf springs is damaged, bent, twisted, etc. If that does not result in a solution, make sure you have tried a new set of batteries. If the problem still exists, the proper recourse is to send it in for repair and/or replacement.
Please visit www.wagnermeters.com for more information.
Latest posts by Wagner Meters (see all)
- Forest Industry Products Support & Training - July 2, 2018
- June 2018 Rapid RH® Newsletter – What’s the Right Moisture Level for Concrete? - June 26, 2018
- May 2018 Rapid RH® Newsletter – What if Your Concrete Won’t Dry? - May 31, 2018