Why do some Rapid RH® 4.0 EX Smart Sensors fit very tightly in the drilled holes?
Rapid RH® 4.0 EX Smart Sensors should be placed in holes drilled with a 4 fluted, carbide-tipped drill bit that falls within the specifications for an ANSI standard ¾” bit. This range is from 0.775” to 0.787”. A tight-fitting Rapid RH® 4.0 EX Smart Sensor is most likely caused by one of two reasons: The first reason could be that your drill bit has “worn” down with use and has lost some of its “diameter”. This will most likely become noticeable after about 60-70 holes drilled. The other reason could be that you have purchased a new drill bit that is on the lower end of the tolerance spectrum for diameter.
Please visit ANATOMY OF A HAMMER DRILL BIT for more information.
Latest posts by Wagner Meters (see all)
- With the Grain News – May 2018 - May 23, 2018
- Surface Testing Group Uses Rapid RH® to Aid Contractors - May 10, 2018
- Video 2 – Relative Humidity Testing for Concrete Floors - May 2, 2018