Probe Placement (Depth) - Printable Version
+- Concrete Moisture Measurement Forums (http://www.wagnermeters.com/concreteforum)
+-- Forum: Concrete (/forumdisplay.php?fid=1)
+--- Forum: Moisture Measurement in Concrete (/forumdisplay.php?fid=2)
+--- Thread: Probe Placement (Depth) (/showthread.php?tid=40)
Probe Placement (Depth) - CDougherty - 01-14-2010 01:11 PM
ASTM F2170 states that the RH probe should read the RH at 40% of the total depth of the slab (or 20% for a slab drying on two sides). While the standard probes are perfect for placement in the 1-1/2"-1-3/4" depth range to measure RH in a 4" slab, we often run into much thicker slabs. Are there other probes, or a longer reader to be inserted into a deeper hole, in order to measure the RH at 40% depth in an 8" or 12" thick slab? If not, how do you propose measuring the RH accurately in such a slab?
Thanks for the help.
RE: Probe Placement (Depth) - Blake McFee - 01-15-2010 03:34 PM
have you seen the Rapid RH long reader? it's 5" long and I've found it to be the easiest to use on the thicker slabs.
RE: Probe Placement (Depth) - CC Solutions - 01-15-2010 09:43 PM
Yes, the longer reader is so much nicer than the original short one.
In the 'old' days, we'd get the little reader down in the hole and lock it into the sensor using a stiff wire with a little bend on the end. You could still do that!