Bad Sensor - Printable Version
+- Concrete Moisture Measurement Forums (http://www.wagnermeters.com/concreteforum)
+-- Forum: Concrete (/forumdisplay.php?fid=1)
+--- Forum: Moisture Measurement in Concrete (/forumdisplay.php?fid=2)
+--- Thread: Bad Sensor (/showthread.php?tid=346)
RE: Bad Sensor - Ernesto - 12-08-2012 08:32 AM
(12-07-2012 07:13 PM)CC Solutions Wrote: I believe with E-96 they knew what they wanted to test for, and then used the product they felt would give the best results.
You said that twice now, throwing salt on the ground. According to Howard that was not the case. The CaCl crystals were in a small cup above a hole which was covered.
He seriously doubts the E-96 of today was the original draft, version, test methodology. I think they did it because of the same reasons you stated above. The 1869 was using CaCl and so they started using it as well. It uses the lbs formula also.
Here's another question. Which ASTM came first, the 1869 or E-96?
RE: Bad Sensor - Ernesto - 12-10-2012 04:48 PM
So where are all the experts on this? No comment? Not so surprising.